Duodenal perforation by an inferior vena cava filter with staphylococcal bacteremia: a case report
© Pokharel et al. 2016
Received: 1 October 2015
Accepted: 17 April 2016
Published: 4 May 2016
The Erratum to this article has been published in Journal of Medical Case Reports 2016 10:186
Inferior vena cava filter complications can range from dislodgement to perforation. Patients who present with concomitant bacteremia have rarely been reported. Persistent bacteremia usually results from direct bacterial seeding from a source other than perforation of surrounding viscus. It is unclear if the risk of perforation is higher in patients who are bacteremic due to other causes.
We report an interesting case of a 67-year-old white woman who presented with fever, chills, and right upper quadrant abdominal pain. Her blood cultures were positive for methicillin-sensitive Staphylococcus aureus with no obvious source. Upon further investigation, she was found to have an inferior vena cava filter perforating her duodenum. The cause of her abdominal pain was explained by the inferior vena cava filter penetrating the duodenum; however, the source of bacteremia could not be ascertained. The inferior vena cava filter was removed successfully, and she was discharged on an intravenous antibiotic. Her symptoms resolved soon after the filter was removed.
The use of inferior vena cava filters has increased significantly in recent years. This is likely due to their wider availability and safer placement techniques. With increasing use, the complications arising from these filters have been on the rise as well. It is very important for clinicians to be aware of these complications to avoid delays in diagnosis and patient care.
KeywordsPerforation Bacteremia Inferior vena cava
Inferior vena cava (IVC) filters are commonly used for the prevention of pulmonary embolisms when anticoagulation is contraindicated or ineffective [1, 2]. Duodenal perforation has been reported as one of the late complications of IVC filter placement. It is exceedingly rare, with only 25 cases reported in a systemic review published in 2012 . We report a unique case of a patient with duodenal perforation by an IVC filter with concomitant methicillin-sensitive staphylococcal bacteremia.
The cause of the patient’s abdominal pain was explained by the IVC filter penetrating the duodenum; however, the source of bacteremia could not be ascertained. The duodenum is a sterile part of the gastrointestinal tract and was unlikely to be the source of bacteremia. It was unclear if the bacteremia was a mere coincidence or whether it contributed to this complication by any means. Hopefully, more cases such as this one will clarify this in the future. A vascular surgeon and an interventional radiologist were consulted. The patient underwent successful retrieval of the IVC filter using a superior vena caval approach with a minimally invasive technique. She experienced no complications. She was discharged on an intravenous antibiotic. Anticoagulation was resumed upon discharge.
An IVC filter provides a mechanical barrier that prevents pulmonary embolisms originating in the veins of the lower extremities, pelvis, and IVC . The indications for filter placement include patients with venous thromboembolism who have contraindications to anticoagulation, those with recurrent pulmonary embolisms while being adequately anticoagulated, and patients who develop complications due to anticoagulation [1, 2]. IVC filters have been used since the early 1970s for the treatment of venous thromboembolic disease, and retrievable filters have been used increasingly since their introduction in 2001 .
Permanent filters cannot be removed or repositioned. Retrieval of the temporary IVC filter is often straightforward and can be done with a high degree of success, ranging from 93 % to 100 % [5, 6]. Removal within 30 days is typical, but successful filter removal more than 1 year after implant has been reported. Kwok et al. described a combined jugular and femoral approach for retrieving an embedded filter .
Caval perforation is a well-known complication of various types of IVC filters, and, while it occurs up to 40 % of cases, it is thought to be largely asymptomatic [8–10]. Multiple case reports highlight the range of potential complications once the integrity of the vena cava has been breached. These complications include upper and lower gastrointestinal bleed, aortic and vertebral erosion, ureteric erosion, and aortoduodenal fistula [1, 3, 4, 9–11]. However, to date, only one case of staphylococcal bacteremia secondary to an infected IVC filter has been reported. This case occurred in an intravenous drug user who presented with multiple abscesses . Despite the obvious benefits of retrievable filters, studies unfortunately suggest that the retrieval rates of temporary or retrievable filters are quite low and seldom exceed 20 % in most series [1, 2, 5].
Duodenal perforation by an IVC filter is an uncommon complication. The concomitant bacteremia found in our patient was more likely coincidental than causal. Once a patient is persistently bacteremic, there is a significant risk of the IVC filter becoming infected. It should be taken out unless there is an absolute contraindication not to do so. It is unclear if bacteremia from another source will make someone vulnerable to this complication. The incidence of duodenal perforation is rising with the increasing use of IVC filters and more readily available diagnostic tools. Physicians should have a high degree of clinical suspicion in someone presenting with persistent abdominal pain despite normal routine evaluations.
Written informed consent was obtained from the patient for publication of this case report and any accompanying images. A copy of the written consent is available for review by the Editor-in-Chief of this journal.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Becker DM, Philbrick JT, Selby JB. Inferior vena cava filters: indications, safety, effectiveness. Arch Intern Med. 1992;152:185–94.View ArticleGoogle Scholar
- Rectenwald JE. Vena cava filters: uses and abuses. Semin Vasc Surg. 2005;18:166–75.View ArticlePubMedGoogle Scholar
- Ward WH, Donahue DR, Platz TA, Scibelli CD. Duodenal penetration of an inferior vena cava filter: case report and literature review. Vascular. 2013;21(6):386–90.View ArticlePubMedGoogle Scholar
- Malgor RD, Labropoulos N. A systematic review of symptomatic duodenal perforation by inferior vena cava filters. J Vasc Surg. 2012;55(3):856–61. e3.View ArticlePubMedGoogle Scholar
- Asch MR. Initial experience in humans with a new retrievable inferior vena cava filter. Radiology. 2002;225:835–44.View ArticlePubMedGoogle Scholar
- Grande WJ, Trerotola SO, Reilly PM, et al. Experience with the recovery filter as a retrievable inferior vena cava filter. J Vasc Interv Radiol. 2005;16:1189–93.View ArticlePubMedGoogle Scholar
- Kwok PC, Wong WK, Siu KW, Lai AK, Chan SC. Difficult retrieval of a retrievable inferior vena cava filter placed in an inverted orientation. J Vasc Interv Radiol. 2006;17:153–5.View ArticlePubMedGoogle Scholar
- Miyahara T, Miyata T, Shigematsu K, Deguchi J, Kimura H, Ishii S, et al. Clinical outcome and complications of temporary inferior vena cava filter placement. J Vasc Surg. 2006;44:620–4.View ArticlePubMedGoogle Scholar
- Sadaf A, Rasuli P, Olivier A, Hadziomerovic A, French GJ, Aquino J, et al. Significant caval perforation by the Celect inferior vena cava filter: attributable to filter design? J Vasc Interv Radiol. 2007;18:1447–50.View ArticlePubMedGoogle Scholar
- Van Ha TG. Complications of inferior vena caval filters. Semin Interv Radiol. 2006;23(2):150–5.View ArticleGoogle Scholar
- Assifi MM, Bagameri G, Dimuzio PJ, Eisenberg JA. Management of infected caval filter with simultaneous aortic pseudoaneurysm and retroperitoneal perforation: a case report and literature review. Vascular. 2012;20(4):225–8.View ArticlePubMedGoogle Scholar