- Case report
- Open Access
Set screw fracture with cage dislocation after two-level transforaminal lumbar interbody fusion (TLIF): a case report
Journal of Medical Case Reports volume 9, Article number: 22 (2015)
Transforaminal lumbar interbody fusion is a popular procedure used to achieve spondylodesis in patients with degenerative lumbar spinal diseases. We present a rare case of a patient with a set screw fracture with cage dislocation after an open transforaminal lumbar interbody fusion procedure. To the best of our knowledge, this case is the first of its kind to be reported.
A 44-year-old Caucasian woman attended a follow-up appointment at our hospital 3 months after treatment for second-degree lumbar spondylolisthesis (L4/L5) and osteochondrosis (L5/S1) with transforaminal lumbar interbody fusion and dorsal spondylodesis. She complained of severe leg pain on the left side. Her physical examination revealed a normal neurological status, except for paresthesia of the entire left lower limb and at the ball of the left foot. Radiological imaging showed breaking of the set screws with cage dislocation. Surgical revision was then performed with exchange of the whole dorsal instrumentation and the dislocated cage. Six weeks post-operatively, the patient was seen again at our clinic without neurological complaints, except for decreased sensitivity on the dorsum of her left foot. The wound healing and radiological follow-up were uneventful.
Hardware-related complications are rarely seen in patients with open transforaminal lumbar interbody fusion, but must be kept in mind and can potentially cause severe neurological deficits.
Transforaminal lumbar interbody fusion (TLIF) is a popular procedure used to achieve spondylodesis in patients with degenerative lumbar spinal diseases . Neurological, wound-healing and hardware problems have been reported in these patients . Nevertheless, hardware problems with implant fractures are a rare complication of the procedure . We present a case of implant fracture with cage dislocation after an open TLIF procedure. To the best of our knowledge, this is a complication after open TLIF that has not been reported in the literature to date.
A 44-year-old Caucasian woman came to our outpatient clinic for a 3-month follow-up appointment after undergoing an open two-level (L4-S1) TLIF procedure with dorsal spondylodesis. She had been treated for spondylolysis with second-degree lumbar spondylolisthesis (L4/L5) and for osteochondrosis (L5/S1) (Figures 1 and 2). At her follow-up visit, she complained of pain in her whole left leg.
A physical examination revealed pain upon palpation of the lumbar spine. The patient’s spinal muscles were tense. Intermittent paresthesia of her left lower limb and constant paresthesia at the ball of the left foot were present. There was no saddle block paresthesia. Her motor function was MRC (Medical Research Council) Scale 5/5 in all muscles of both legs. Her Lasègue and Bragard signs were positive on the left side. Her bilateral patellar and Achilles tendon reflexes were regular, as was her perianal sensitivity and sphincter muscle tone. No urinary or defecation problems were observed. The patient was in good general condition, but overweight (108 kg weight, 170 cm height). She had sustained no trauma. She had been compliant with the given instructions for sitting and working in a manner that is easy on the back. However, she had not lost any body weight as we had advised.
Her past medical history included arterial hypertonus, restless legs syndrome and an axial esophageal hernia. Her past surgical history consisted of a hysterectomy and tonsillectomy, 6 and 31 years ago, respectively. Upon admission to the hospital, she was taking several painkillers, blood pressure medication, a dopamine agonist and a proton pump inhibitor.
Her social history revealed that she had consumed nicotine since she was 14 years old (46.5 pack-years). Her family history and review of her organ systems were non-contributory, except for the above-mentioned diseases.Radiography and computed tomography were performed (Figures 3 and 4). A diagnosis of implant failure with a combination of cage dislocation left dorsally and breaking of the set screws on both sides with dislocation of the rods was made. All polyaxial screws were intact and in place, except for the S1 screws, which had become loose and dislocated dorsally on both sides.
A revision surgery was performed by the same surgical team that had done the primary operation. The patient was placed in prone position. A midline incision was made through the existing scar. Removal of cicatricial tissue was performed until all of the screws could be exposed. The S1 polyaxial screws were loose and dislocated on both sides. Metallosis was observed in the surrounding tissue. The S1 set screws were then inspected and removed. They showed a breaking of the worm. The other polyaxial screws were in place and intact. The L5 nerve root was prepared, followed by exposure of the dura. From there, preparation toward the cage in the L5/S1 intervertebral disc space was done. Palpation showed that it had become completely loose. The cage was removed. Next, expansion of the intervertebral disc space, rasping of the edges and insertion of a new cage was done. After that, the whole dorsal instrumentation system was exchanged. Pedicle screws with a 0.5-mm-larger outer diameter than those previously used were placed. Finally, the wound was closed layerwise, and two drains were placed. No intra-operative complications occurred.A post-operative X-ray showed a regular position of the implants (Figure 5). The drains were removed on the second post-operative day. The wound healing was primary and without signs of infection. The patient’s neurological complaints (pain and paresthesia) disappeared completely. Again, physiotherapists instructed the patient to sit and work in a manner that is easy on the back, and she was advised to lose weight and quit smoking. Twelve days after the operation, the patient was discharged to home. From there, she went into a rehabilitation clinic. No peri-operative complications occurred.Six weeks post-operatively, the patient was seen at our outpatient clinic for a follow-up examination (Figure 6). She complained of back pain without radiation to the legs, which was controllable with pain medication. In addition to that, she complained that the dorsum of her left foot had decreased sensitivity. A physical examination revealed a regular scarring of the wound. Paresthesia was present on the dorsum of her left foot. Apart from that, sensitivity was the same bilaterally. Her motor function was MRC (Medical Research Council) Scale 5/5 in all muscles of the legs bilaterally, and she had regular reflexes. Her Lasègue and Bragard signs were negative bilaterally. No post-operative complications were observed.
To identify similar cases, we reviewed the online databases PubMed, MEDLINE, CareLit and MEDPILOT for the keywords "TLIF," "transforaminal lumbar interbody fusion" and "complication." The search revealed 244 publications. We excluded 204 studies that covered inappropriate topics. This left us with 40 studies with comparable patients and treatments. Implant-related complications were found in 22 of these studies and were present in 76 cases (Table 1) [1–22].
Of these hardware-related complications, 75 were further specified and 1 was unspecified. There were 31 cases with cage or pedicle screw malpositioning, 30 with cage migration, 3 with cage subsidence, 9 with pedicle screw loosening or backing out and 2 with a rod fracture. However, none of these cases were comparable to the one described in the present report, which uniquely consisted of loosening of the polyaxial sacral screws, fracturing of the set screws and cage dislocation.
Although our report provides no causal explanation for this kind of implant failure, we assume that the size and placement of the implant did matter in our patient. Furthermore, we believe that the dorsal fixation device may have been too small for our patient’s biomechanical needs. Cho and coworkers reported that pull-out strength of the pedicle screws increased with the outer diameter of the screws . Therefore, we assume that our primary choice to use pedicle screws with a smaller outer diameter led to early loosening of the sacral screws in our patient. Only after exchanging the primary system for a larger one did we achieve good biomechanical stability.
The broken set screws make the present case unique. We could not identify any matching cases that have been reported to date. However, in a similar report, Agrawal described a case of nut loosening, and thus a loosened set screw after dorsal instrumentation, but without breakage of the worm . Moreover, nut loosening was reported by Davne and Myers for dorsal instrumentation and posterior lumbar interbody fusion with a variable screw placement system, which uses a plate over which screws are placed into the bone . In their large series of 486 patients, they attributed a certain amount of surgical inexperience to most cases of nut loosening. In our case, we assume that it was not inexperience of the operating surgeons that led to breakage of these screws. However, we acknowledge that, in general, malpositioning of the set screws or flaws in their production can lead to a certain predisposition for the worm to break.
With regard to fixation of the pedicle screws, Ha and coworkers showed that side fixation of the pedicle screws led to greater stability than head fixation with a set screw . We were unable to identify any literature addressing whether an increase of the contact face of the set screws also leads to greater stability. Nevertheless, in our patient, the exchange of the internal fixation system led to the implantation of a set screw with a larger contact face. Whether an increase of the contact face of the set screws leads to greater stability remains unclear.
Cage migration typically occurs dorsally, as it did in our patient . It is also associated with cage placement in a central position as compared to cages placed in anterior positions . In our patient, the migrated cage between L5 and S1 lay rather dorsally. This may have also predisposed the cage to migrate in this direction.
With regard to our patient’s obesity, the available literature shows no difference in operative time, length of hospitalization or complication rates in general between obese and non-obese patients . Nevertheless, this remains a controversial topic, and therefore obesity must be kept in mind as a possible negative factor.
The fact that our patient has many pack-years of smoking is unlikely to have contributed to her outcome. Our review of the literature showed that negative effects of smoking in TLIF have not been observed and can probably be excluded as a factor related to poor outcome [1, 2, 20].
The complications we describe in this report are particularly of interest for spinal surgeons. To the best of our knowledge, this report is currently the only available one of its kind in the literature.
The following take-home messages must be taken into account. (1) Hardware-related problems are a rare complication in open TLIF, but must be kept in mind. (2) Hardware-related problems must be mentioned when obtaining informed consent from a patient prior to performing open TLIF. (3) Hardware-related complications can potentially cause severe neurological deficits. (4) Lifestyle modifications should be advised for all patients, if appropriate.
Written informed consent was obtained from the patient for publication of this case report and any accompanying images. A copy of the written consent is available for review by the Editor-in-Chief of this journal.
Transforaminal lumbar interbody fusion.
Tormenti MJ, Maserati MB, Bonfield CM, Gerszten PC, Moossy JJ, Kanter AS, et al.: Perioperative surgical complications of transforaminal lumbar interbody fusion: a single-center experience.J Neurosurg Spine 2012, 16:44–50. 10.3171/2011.9.SPINE11373
Takahashi T, Hanakita J, Minami M, Kitahama Y, Kuraishi K, Watanabe M, Takeshima Y, Uesaka T: Clinical outcomes and adverse events following transforaminal interbody fusion for lumbar degenerative spondylolisthesis in elderly patients.Neurol Med Chir 2011, 51:829–35. 10.2176/nmc.51.829
Burneikiene S, Nelson EL, Mason A, Rajpal S, Serxner B, Villavicencio AT: Complications in patients undergoing combined transforaminal lumbar interbody fusion and posterior instrumentation with deformity correction for degenerative scoliosis and spinal stenosis.Surg Neurol Int 2012, 3:25. 10.4103/2152-7806.92933
Castro FP Jr: Role of activated growth factors in lumbar spinal fusions.J Spinal Disord Tech 2004, 17:380–4. 10.1097/01.bsd.0000110342.54707.19
Hackenberg L, Halm H, Bullmann V, Vieth V, Schneider M, Liljenqvist U: Transforaminal lumbar interbody fusion: a safe technique with satisfactory three to five year results.Eur Spine J 2005, 14:551–8. 10.1007/s00586-004-0830-1
Lauber S, Schulte TL, Liljenqvist U, Halm H, Hackenberg L: Clinical and radiologic 2–4-year results of transforaminal lumbar interbody fusion in degenerative and isthmic spondylolisthesis grades 1 and 2.Spine (Phila Pa 1976) 2006, 31:1693–8. 10.1097/01.brs.0000224530.08481.4e
Weiner BK, Nguyen HV, Hazard SW: Transforaminal lumbar interbody fusion: an independent assessment of outcomes in a difficult patient population.Med Sci Monit 2006, 12:CR99–102.
Taneichi H, Suda K, Kajino T, Matsumura A, Moridaira H, Kaneda K: Unilateral transforaminal lumbar interbody fusion and bilateral anterior-column fixation with two Brantigan I/F cages per level: clinical outcomes during a minimum 2-year follow-up period.J Neurosurg Spine 2006, 4:198–2015. 10.3171/spi.2006.4.3.198
Yan D, Pei F, Li J, Soo C: Comparative study of PILF and TLIF treatment in adult degenerative spondylolisthesis.Eur Spine J 2008, 17:1311–6. 10.1007/s00586-008-0739-1
Xu H, Tang H, Li Z: Surgical treatment of adult degenerative spondylolisthesis by instrumented transforaminal lumbar interbody fusion in the Han nationality.J Neurosurg Spine 2009, 10:496–9. 10.3171/2009.2.SPINE08623
Faundez AA, Schwender JD, Safriel Y, Gilbert TJ, Mehbod AA, Denis F, et al.: Clinical and radiological outcome of anterior-posterior fusion versus transforaminal lumbar interbody fusion for symptomatic disc degeneration: a retrospective comparative study of 133 patients.Eur Spine J 2009, 18:203–11. 10.1007/s00586-008-0845-0
Gong K, Wang Z, Luo Z: Reduction and transforaminal lumbar interbody fusion with posterior fixation versus transsacral cage fusion in situ with posterior fixation in the treatment of Grade 2 adult isthmic spondylolisthesis in the lumbosacral spine.J Neurosurg Spine 2010, 13:394–400. 10.3171/2010.3.SPINE09560
Fujibayashi S, Neo M, Takemoto M, Ota M, Nakamura T: Paraspinal-approach transforaminal lumbar interbody fusion for the treatment of lumbar foraminal stenosis.J Neurosurg Spine 2010, 13:500–8. 10.3171/2010.4.SPINE09691
Wang J, Zhou Y, Zhang ZF, Li CQ, Zheng WJ, Liu J: Comparison of one-level minimally invasive and open transforaminal lumbar interbody fusion in degenerative and isthmic spondylolisthesis grades 1 and 2.Eur Spine J 2010, 19:1780–4. 10.1007/s00586-010-1404-z
Mura PP, Costaglioli M, Piredda M, Caboni S, Casula S: TLIF for symptomatic disc degeneration: a retrospective study of 100 patients.Eur Spine J 2011, 20:57–60. 10.1007/s00586-011-1761-2
Aoki Y, Yamagata M, Ikeda Y, Nakajima F, Ohtori S, Nakagawa K, Nakajima A, Toyone T, Orita S, Takahashi K: A prospective randomized controlled study comparing transforaminal lumbar interbody fusion techniques for degenerative spondylolisthesis: unilateral pedicle screw and 1 cage versus bilateral pedicle screws and 2 cages.J Neurosurg Spine 2012, 17:153–9. 10.3171/2012.5.SPINE111044
Kim PD, Baron EM, Levesque M: Extrusion of expandable stacked interbody device for lumbar fusion: case report of a complication.Spine (Phila Pa 1976) 2012, 37:E1155–8. 10.1097/BRS.0b013e318257f14d
Lee KH, Yue WM, Yeo W, Soeharno H, Tan SB: Clinical and radiological outcomes of open versus minimally invasive transforaminal lumbar interbody fusion.Eur Spine J 2012, 21:2265–70. 10.1007/s00586-012-2281-4
Høy K, Bünger C, Niederman B, Helmig P, Hansen ES, Li H, Andersen T: Transforaminal lumbar interbody fusion (TLIF) versus posterolateral instrumented fusion (PLF) in degenerative lumbar disorders: a randomized clinical trial with 2-year follow-up.Eur Spine J 2013, 22:2022–9. 10.1007/s00586-013-2760-2
Lau D, Khan A, Terman SW, Yee T, La Marca F, Park P: Comparison of perioperative outcomes following open versus minimally invasive transforaminal lumbar interbody fusion in obese patients.Neurosurg Focus 2013, 35:E10.
Seng C, Siddiqui MA, Wong KPL, Zhang K, Yeo W, Tan SB, Yue WM: Five-year outcomes of minimally invasive versus open transforaminal lumbar interbody fusion: a matched-pair comparison study.Spine (Phila Pa 1976) 2013, 38:2049–55. 10.1097/BRS.0b013e3182a8212d
Zhang K, Sun W, Zhao C, Li H, Ding W, Xie Y, Sun X, Zhao J: Unilateral versus bilateral instrumented transforaminal lumbar interbody fusion in two-level degenerative lumbar disorders: a prospective randomised study.Int Orthop 2014, 38:111–6. 10.1007/s00264-013-2026-y
Cho W, Cho SK, Wu C: The biomechanics of pedicle screw-based instrumentation.J Bone Joint Surg (Br) 2010, 92:1061–5.
Agrawal A: Pedicle screw nut loosening: potentially avoidable causes of spine instrumentation failure.Asian Spine J 2014, 8:224–6. 10.4184/asj.2014.8.2.224
Davne SH, Myers DL: Complications of lumbar spinal fusion with transpedicular instrumentation.Spine (Phila Pa 1976) 1992,17(6 Suppl):S184–9.
Ha KY, Hwang SC, Hwang TH: Biomechanical stability according to different configurations of screws and rods.J Spinal Disord Tech 2013, 26:155–60. 10.1097/BSD.0b013e31823ba058
Fukuta S, Miyamoto K, Hosoe H, Shimizu K: Kidney-type intervertebral spacers should be located anteriorly in cantilever transforaminal lumbar interbody fusion: analyses of risk factors for spacer subsidence for a minimum of 2 years.J Spinal Disord Tech 2011, 24:189–95. 10.1097/BSD.0b013e3181e9f249
Wang J, Zhou Y, Zhang ZF, Li CQ, Zheng WJ, Liu J: Comparison of the clinical outcome in overweight or obese patients after minimally invasive versus open transforaminal lumbar interbody fusion.J Spinal Disord Tech 2014, 27:202–6. 10.1097/BSD.0b013e31825d68ac
We thank American manuscript editors for revising the language, grammar and style of the manuscript.
The authors declare that they have no competing interests.
PL, AH, IH, SH, HK and SL participated in the conception and design of this report and were involved in drafting and revising the manuscript. PL performed the final revisions of the manuscript. All authors read and approved the final manuscript.
About this article
Cite this article
Leute, P.J.F., Hammad, A., Hoffmann, I. et al. Set screw fracture with cage dislocation after two-level transforaminal lumbar interbody fusion (TLIF): a case report. J Med Case Reports 9, 22 (2015). https://doi.org/10.1186/1752-1947-9-22
- Set screw
- Transforaminal lumbar interbody fusion