- Case report
- Open Access
Right subclavian vein catheterism complication due to a 'foreign body': a case report
© Sidiropoulou et al; licensee BioMed Central Ltd. 2010
Received: 22 October 2009
Accepted: 19 October 2010
Published: 19 October 2010
Central venous access devices are widely used in hospital practice. Complications associated with their use are well described and reviewed. In this paper, we report a former complication that in turn created a new complication during a standardized procedure.
We report the case of an 81-year-old Caucasian woman requiring total parenteral nutrition due to a high-debt enterocutaneous fistula. In a previous right subclavian catheterization a fragmentation of the tip of the catheter, probably not recognized at the time, provoked an extrinsic compression of the vessel.
Fragmentation of a central venous catheter is a possible complication of catheterization and can be missed. Control of a catheter is imperative after its removal, even if not always practiced.
Central venous access devices are widely used for the administration of antibiotics and chemotherapeutic drugs, total parenteral nutrition, providing high-flow access for hemodialysis and plasmapheresis, and central venous pressure monitoring. In many cases the same patient will undergo this procedure on more than one occasion, leading to an increase in the possibility of complications.
Central venous catheterization has multiple advantages, for example the reduction of irritation and thrombosis of smaller peripheral veins, the avoidance of peripheral phlebitis and scarring, and a much better patient tolerance. The immediate complications are insertion site bleeding, pneumothorax and hemothorax, arterial puncture, displacement of the catheter and fragmentation of the catheter [1, 2]. Late complications can be catheter infection, surgical site infection, occlusion, endocarditis, and valve embolism .
An 81-year-old Portuguese Caucasian woman, in the immediate post-operative period for incisional relapsing hernia, developed an enterocutaneous, high debit fistula requiring total parenteral nutrition.
Our patient presented no other major medical comorbidity. Her surgical history consisted of seven abdominal operations. The first was a total hysterectomy with bilateral adnexectomy by a midline abdominal incision (1979). Secondly, two years later she developed an incisional hernia and was submitted to a herniorraphy (1981). Her third operation was a laparotomy for intestinal occlusion due to adhesions (1984). Then, three years later she developed a new incisional hernia that was corrected by hernioplasty (1991). Because of surgical site infection, the prosthesis had to be removed and replaced (1991). After a new episode of intestinal occlusion, this time with necrosis, a segmental resection of ileum (2004) was performed, after which she presented with a recurrence of the incisional hernia and was operated on again by hernioplasty (2008). There was a new recurrence of the incisional hernia one year later and an application of biological prosthesis was completed (2009).
The surgical team had no data about the intestinal occlusions and the following resection operations that were reported later by one of the daughters of our patient, who lived abroad. No detailed medical reports had been presented, and it seemed that some of the operations had been performed in another hospital during occasional stays of our patient at her daughter's home abroad.
The first approach to the central venous catheterization was made through the right subclavian vein, following the Seldinger technique. During the introduction of the wire, resistance was encountered, so the surgeon extracted the guide and reattempted introduction. During this second attempt the arterial vessel was accidentally punctured and local compression was applied to successfully stop the bleeding.
A new approach was attempted with the same catheter on the left subclavian vein, which was successful and without complications. A control chest X-ray was ordered and showed the correct positioning of the catheter.
We presumed that the foreign body detected was a central venous catheter tip. This probably fragmented during the extraction of the device placed in 2004 when our patient underwent intestinal resection, and later migrated to the interstitial space between the vessels. However, we have no means to confirm our theory. The only certainty we had is that it is not a complication of our procedure since we did not change the catheter during its replacement and our patient's CT scan results showed two catheter tips.
We did not take any additional surgical or interventional measures, since our patient was asymptomatic and the 'embolus' was fixed in the interstitial space. Nevertheless, our patient has continued taking enoxaparin 20 mg subcutaneous daily in order to prevent any thromboembolic complications .
Taking of a thorough medical history is extremely important for the safe and successful management of a patient, but it is not always possible to obtain. Central venous catheterization complications can be misdiagnosed by the time they occur. When rare difficulties during catheter placement occur, the possibility that relevant data could be missing from a patient's clinical history with regard to previous complications should be considered. It is also good practice to check catheters and perform microbiological cultures of the tip. An ultrasound-guided catheter insertion could possibly have detected, in real time, the extrinsic compression and further manipulation could have been avoided.
Written informed consent was obtained from the patient for publication of this case report and any accompanying images. A copy of the written consent is available for review by the Editor-in-Chief of this journal.
We thank Pedro Gameiro (Surgery Chief), Ana Cecilia Sousa (Surgery Assistant) and Afonso Janeiro (Surgery Department Director) for their help in managing our patient.
- Marcondes C, Biojone C, Cherri J, Moryia T, Piccinato C: Early and late complications in long term central venous access. Analysis of 66 implants [in Portugese]. Acta Cir Bras. 2000, 15: 73-75.View ArticleGoogle Scholar
- Seldinger technique. http://www.cvc-partner.com/index.cfm?7F880D3CB24B4A6586A771F17149822C
- Emedicine Health: Venous Access Devices. http://www.emedicinehealth.com/venous_access_devices/page3_em.htm
- Randolph AG, Cook DJ, Gonzales CA, Andrew M: Benefit of heparin in central venous and pulmonary artery catheters: a meta-analysis of randomized controlled trials. Chest. 1998, 113: 165-167. 10.1378/chest.113.1.165.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.