A simple way to distinguish bed clothing contamination in a whole body bone scan: a case report
© Assadi et al; licensee BioMed Central Ltd. 2007
Received: 18 July 2007
Accepted: 05 December 2007
Published: 05 December 2007
Whole body bone scan with Technetium-99m MDP (methylene diphosphate) can detect bony lesions due to altered osteoblastic activity.
Non-physiologic or increased radiotracer uptake in the bony structures of patients with a history of malignant diseases is usually interpreted as being suspicious of bone metastasis. It is extremely important to properly distinguish false positive sites of Tc-99m MDP localization. We present three patients with the same pattern of Tc-99m MDP abnormality in different locations. These scans were all performed on the same day to evaluate possible bone metastases in three patients with breast carcinoma. After careful examination, repeated images revealed bed clothing contamination. This is different from bed contamination by displacement among different patients. It is also different from detector contamination by limited area of involvement where detector contamination appears as a line throughout the total body projection. It can be helpful if a nuclear medicine specialist has a brief look at all scans prior to reporting them. In cases where the same pattern of abnormality is repeated in all images, the possibility of technical error such as bed clothing contamination rather than a pathological abnormality should be borne in mind.
Whole body imaging with Tc-99m MDP is able to detect bone metastasis. After administration of Tc-99m MDP physiologic activity is noted in a fairly homogeneous pattern in bony structures, and this is generally symmetrical. Urinary bladder and kidneys are visualized as normal findings . However, on some occasions, increased activity in the bony or extra-skeletal structures is not due to bone-related pathologies. Contamination is one of the most important causes . We describe three patients with the same pattern of Tc-99m MDP contamination occurring in different places. All of the scans were performed on a same day in patients to exclude possible bone metastasis from breast carcinoma.
Tc-99m MDP whole body scan is a valuable procedure for detecting bone metastasis; however the possibility of false positive results due to technical error or contamination should always be kept in mind. This is particularly true for lesions seen only on a single view . A common source of contamination results from the intravenous injection of the radioisotope. Contamination may result from injection site dose infiltration, a leaking intravenous tube, or from bleeding from the puncture site because of inadequate hemostasis . Also, urinary contamination is a relatively common finding in whole body bone scan (WBBS) .
Sometimes the body or clothing contaminants migrate through perspiration-soaked areas, typically the knees and forearms, contaminating a protective cloth or table, which may result in confounded and misinterpreted final images . Some reports also indicate adverse working conditions, including hot and humid or cramped working environments, can cause contamination . There are several methods to control these factors. For example, decontaminating different anatomical sites on the body, to eliminate or reduce the uptake of radioactivity absorbed into the body, and cleaning and decontaminating dry and wet surface, equipment and clothing.
We observed a similar pattern of abnormality in different places in three referred patients. We considered the possibility of technical error as an explanation for these findings. After careful examination and multiple spot views, bed clothing contamination was confirmed as the cause of these abnormalities. Clothing contamination differs from bed contamination due to the observation of changeable locations in different patients. It is also different from detector contamination by its limited area of involvement since detector contamination appears as a line throughout the whole body on the total body projection .
Our findings suggest that if on brief review of images performed on a same day, the same abnormal pattern of activity is found in different bone scans, technical errors or contaminations rather than actual bone abnormalities should be the first consideration.
Written informed consent was obtained from all three patients for publication of this case report and any accompanying images. Copies of these written consents are available for review by the Editor-in-Chief of this journal.
List of abbreviations used
whole body bone scan
We are indebted to the technologists at our department for data acquisition and other technical support.
- Eggli DF, Tulchinsky M: Normal planar bone scan. Skeletal nuclear medicine. Edited by: Collier BD, Fogelman I, Rosenthall L. 1996, Mosby, USA, 25-26.Google Scholar
- Mosman Elton, Peterson Luke, Hung Joseph, Gibbons Raymond: Practical Method for Reducing Radioactive Contamination Incidents in the Nuclear Cardiology Laboratory. J Nucl Med Technol. 1999, 27: 287-289.PubMedGoogle Scholar
- Kasner DL, Spieth ME: The day of contamination. J Nucl Med Technol. 2003, 31 (1): 21-4.PubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.